out of 29
Post on 01-Dec-2014
Embed Size (px)
- 1. TOTAL LEUKOCYTE COUNT Dr.Gopala Krishnan.B.Yadav
- 2. It is a blood test to measure the number of white blood cells
- 3. It is a blood test to measure the number of white blood cells Normal range category Range / microlitre adults 4000-10,000 birth 10,000-25,000 1-3yr 6000-18,000 4-7yr 6000-15,000 8-12yr 45000-13,500
- 4. Alteration in number and type of leukocytes is often associated with disease process & hence the count is informative
- 5. LEUCOCYTOSIS Infections Malignancy like leukemia Severe emotional or physical stress Pregnancy
- 6. LEUCOPENIA Infections bacterial-(typhoid,paratyphoid) viral(hepatitis) protozoa(malaria) Some cases of leukemia Primary bone marrow depression (Aplastic anaemia) Secondary bone marrow depression (drugs,radiation) anaemia ( megaloblastic)
- 7. LETS COUNT
- 8. SPECIMENS Double oxalated or EDTA blood Capillary blood
- 9. REQUIREMENTS Microscope Hemocytometer Wbc diluting fluid
- 10. HEMOCYTOMETER Hemo: blood Cyto: cell Meter: measurement/counter Thus, it is an instrument used to count the blood cells.
- 11. It includes:a) Neubauers slideb) Cover slipc) RBC pipetted) WBC pipette
- 12. NEUBAUERS SLIDE It is the name given to a thick glass slide . In the centre of the slide, there is an H- shaped groove. On the two sides of the central horizontal bar, there are scales for counting the blood cells The depth of the scales is 1/10mm or 0.1mm.
- 13. NEUBAUERS CHAMBER Neubauers slide with a cover slip over it, is called a Neubauers chamber Four corner squares are meant for WBC counting. Total = 64 small squares
- 14. WBC PIPETTE It has a white bead It has graduations upto mark 11 Size of bulb is smaller Size of lumen is larger
- 15. WBC DILUTING FLUID:It is prepared as follows: a) Glacial acetic acid: 2.0 ml b) 1 % (w/v) gentian violet: 1.0 ml c) Distilled water: 97 ml
- 16. PRINCIPLE The glacial acetic acid lyses the red cells while the gentian violet slightly stains the nuclei of the leukocytes. The blood specimen is diluted 1:20 in a WBC pipette with the diluting fluid. The cells are counted under low power of the micro scope by using a counting chamber. The number of cells in undiluted blood are reported per cu mm (l) of whole blood
- 17. PROCEDURE Draw blood up to 0.5 mark of a WBC pipette. Carefully, wipe excess blood outside the pipette by using cotton. Draw diluting fluid up to 11 mark. Mix the contents in the pipette and after five minutes by discarding few drops, fill the counting chamber and allow the cells to settle for two to three minutes.
- 18. FOCUSING 4X to see the general formation of slide. 10X for WBC counting
- 19. COUNTING RULE Do not count cells touching Bottom line Right line This is to avoid double counting.
- 20. WBC COUNTING
- 21. DILUTION FACTOR0.5 part of blood is mixed in 10 parts of fluidSo, 1 part of blood is in 20 parts of fluidThus, dilution factor for WBC counting is 20.
- 22. CALCULATION OF THE VOLUME OF WBC SQUARES Length of one small square = 1/4mm Width of one small square = 1/4mm Depth of one small square = 1/10mm Volume of one small square = 1/4 x 1/4 x 1/10 = 1/160mm
- 23. WBC COUNTING Total no. of WBCs in 64 smallest squares = X No. of WBCs in one smallest square = X/64 Volume of one smallest square = 1/160mm No. of WBCs in 1/160mm = X/64 No. of WBCs in 1mm = X/64 x 160 No. of WBCs in 20 times diluted blood = X/64 x 160 No. of WBCs in undiluted blood = X/64 x 160 x 20/mm = X x 50/mm
View more >